RDS Crack Propagation Gauges For Determining Crack Propagation in Components
HBM crack propagation gauges are used to determine crack propagation in a component. HBM offers three different types.
Types RDS20 and RDS40 consist of electrically isolated resistors. In this case individual circuits are interrupted as the crack propagates.
Type RDS22 consists of conductor tracks connected in parallel which will tear if the crack extends under the crack propagation gauge. This will gradually increase the electrical resistance of the strain gauge as the crack continues to extend.
This change in resistance can be measured using a resistance meter or strain gauge amplifier.
Connecting a Crack Propagation Gauge
There are two different types of crack propagation gauges: Type RDS22 consists of conductor tracks connected in parallel which will tear if the crack extends under the crack propagation gauge.
This will gradually increase the electrical resistance of the strain gauge as the crack continues to extend. Types RDS20 and RDS40 consist of electrically isolated conductor tracks, i.e. as the crack extends, individual circuits will be interrupted. If the circuits are contacted individually, the direction in which the gap is extending can be detected.
The easiest way to detect the signals of crack propagation gauges (RDS) is with a resistance measurement. Direct resistance measurements of this type can be taken with many amplifiers from HBM (e.g. MGCplus or Spider8 with the appropriate modules).
The resulting resistance (R) of the RDS is dependent on the number of torn links and can easily be calculated. In this case n identical resistors (Ri) are connected in parallel:
If a grid line is interrupted, this is described by
However, the measurement can be performed just as well with a measuring amplifier for SG measurements. The circuit diagram sketch shows how the RDS has to be complemented to obtain a resistance change that is inside the measuring range of an amplifier for strain gauge quarter bridges.
To minimize the effects of temperature, temperature-stable fixed resistors or SGs should be used as completion resistors. Greater sensitivity can be obtained by selecting a parallel resistor with a higher resistance value.
HBM의 표준 프로그램 상의 어떤 스트레인게이지로도 충분히 다루지 못한 특별한 요건을 가지고 계십니까? 현재 사용하고 계신 제품에 대한 등가 스트레인게이지를 찾고 계십니까? 자체 스트레인게이지를 설계해 본 적이 있으십니까? HBM으로 연락 주시면 여러분의 요건에 따라 맞춤형 스트레인게이지를 생산해 드리겠습니다. 최소 수량은 20패키지입니다.
|제목 / 설명||언어||제품||언어||종류|
|Dehnungsmessstreifen - Broschüre||German|
|Strain Gauges - Brochure||English|
|Download: Technical Article / White Paper|
|Optical Fiber Sensors vs. Conventional Electrical Strain Gauges for Infrastructure Monitoring Applications||English|