Somat eDAQ Isolated Thermocouple Layer
The Somat eDAQ Isolated Thermocouple Layer (EITB) measures temperatures on eight channels of isolated thermocouple signal conditioning. Thermocouples come in several styles of dissimilar metals, or calibrations; and the EITB is available in the four most common calibration types (J, K, T and E). Each channel has individual cold junction compensation with isolation to 500V.
All eight channels on any one layer must be of a single calibration type.
Size: 9.2"W x 9.7"L x 1.3"H (23cmW x 25cmL x 3.3cmH)
Weight: 4.4lbs (2.0kg)
Temperature: -20° to 65° C
Connector: Female Miniature Spade Thermocouple
Accuracy: +/- 0.5° C
Isolation: Channels Independently Isolated to 500V
Resolution: 16-bit ADC
Cold Junction Compensation: One per channel
Sample Rates: Independently Set, 0.1Hz-5Hz
Filtering: Guard Filters are used before the ADC on each channel, followed by digital filtering.
Type Range K Type -200° C to 1372° C T Type -200° C to 400° C E Type -200° C to 1000° C J Type -200° C to 1200° C
(8) J-SMP-x-M Male Miniature Spade Connectors (x denotes the calibration type J,K,T,E)
Part Number Code
The following number scheme is used to identify the layer configuration.
EITB - X
eDAQ Layer EITB
J = Thermocouple Calibration Type J
K = Thermocouple Calibration Type K
T = Thermocouple Calibration Type T
E = Thermocouple Calibration Type E