The Proper Installation of Strain Gauges on Concrete Surfaces
Installing strain gauges (SG) on concrete surfaces does not have to be difficult. Bear in mind three key criteria and the rest will be a cakewalk. Read on to find out what these criteria are and what you have to take into account.
The choice of the right SG depends on the measuring grid length, mainly because it has a decisive impact on the result of the measurement. The SG determines the arithmetic mean of the strain conditions existing under its measuring grid. This factor needs to be taken into account to answer the question regarding the optimum measuring grid length. When a short measuring grid is used to measure the strain on inhomogeneous material such as concrete, partial strains in pebbles or individual sections of the hardened cement paste would be measured. In this case, it is essential to comply with the recommendation to use a measuring grid length five times as long the largest aggregate grain size. Taking this into consideration, HBM provides strain gauges with up to 150mm measuring grid length (for instance HBM 1-LY41-150/120)
Which adhesive is particularly suitable?
Concrete requires a strong, pore-filling adhesive that forms reliable bonds even if some amount of residual moisture is still present in the concrete. The superglue X60 produced by HBM has proven its mettle in such situations. Direct bonding of SG with concrete is only recommended for dry parts. If residual moisture is present, it is recommended that a thin aluminum foil be applied first as a barrier layer, and the SG be installed on top of the foil.
Zero-point related measurements are only possible on concrete if the sample is completely dry or if the moisture content remains constant during the measurement. Concrete is known to shrink or swell if the moisture content in it undergoes changes. Therefore, why zero-point related measurements are usually only possible over relatively short periods of time, unless an equivalent, unloaded object is available for compensation. The rules for metal installation apply to installations on reinforced concrete. You only need to make sure that the strain gauge is protected accordingly. Special attention needs to be paid to the mechanical protection of the SG to ensure that pouring the concrete will not result in damage to the strain gauge installation.
WTX120: Industrielles, Eichfähiges Wägeterminal
An das industrielle, auf Hutschienen montierbare Hochleistungs-Wägeterminal WTX120 lassen sich alle analogen DMS-basierten Wägezellen anschließen. Der Indikator ist optimal für Abfüll-, Dosier- und Sortieranlagen und -maschinen geeignet und bietet alles, was Sie von einer modernen Wägeelektronik erwarten können: Dank industrieller Ethernet- und Feldbus-Schnittstellen kann die gesamte Messkette in die gängigen Leitsysteme integriert werden.
Display, Bedientasten und Klartext-Menüs ermöglichen die einfache und komfortable Bedienung vor Ort. Als zusätzliches Plus unterstützt die intuitive Wägesoftware PanelX von HBM Sie bei Inbetriebnahme und Serviceeinsätzen.
What must be considered when preparing the measuring point?
The preparation of concrete bonding surfaces is tailored to the use of X60 superglue and is generally simpler than in the case of metal installations. It is necessary to determine whether the concrete was cast with oiled or dry molds. In the case of the former, the oil-soaked layer must be removed with a grindstone. De-greasing with the help of solvents is not recommended, as the solvent and the dissolved oil will only penetrate deeper into the concrete. If dry molds were used, simply remove the concrete laitance until you reach solid concrete. Grinding is also recommended here to obtain as flat a surface as possible.
Further, carefully blow off the grinding dust with an air pump or with oil and water-free compressed air. Thoroughly seal the pores with X60 adhesive, filling them completely, not just superficially. Smoothen the bonding areas, do not apply a layer. After approximately 30 minutes, a thin aluminum foil can be applied as a barrier layer or, under certain circumstances, the SG can be attached.