Real plug and measure:
- Choose from off-the-shelf sensors, amplifiers, cables, and connectors
- Plug everything in
- Get started
TEDS (Transducer Electronic Data Sheet) allows you to just plug in and get started. The transducer characteristics required to set up the measurement chain are stored in the TEDS in the form of an electronic data sheet, which is read by the amplifier. This significantly simplifies the handling of sensors.
Just choose from off-the-shelf sensors, amplifiers, cables, and connectors and configure according to your needs.
The TEDS module can either be integrated in the sensor cable or in the transducer plug.
The transducer characteristics required to set up the measurement chain are stored in the TEDS in the form of an electronic data sheet, which is read by the amplifier. All data is stored in so-called templates in the TEDS chip. These templates can be imagined as tables, where the sensor parameters are listed. The amplifier reads the information and the measurement chain can be set up automatically in no time.
Besides the manufacturer, the serial number and other information, the sensor’s sensitivity, which is the most important information, is, of course, stored in the TEDS
The sensitivity recorded in the test record is stored in all TEDS sensors from HBM coming with a test record which, in general, is included as standard, except for KMR force washers and SLB strain transducers
If a calibration is ordered (e.g. according to the ISO 376 standard), the calibration result can be optionally stored in the TEDS instead of the value recorded in the test record.
Yes, the sensor’s sensitivity (rated output) can be overwritten with new data, for example, in case of recalibration.
catman software also enables the information in the sensor database to be used to easily overwrite a TEDS.
HBM amplifiers, for example QuantumX or MGCplus, allow for writing to a sensor
No, TEDS is based on the IEEE 1451.4 standard which is recognized by many manufacturers.
|TEDS - Data Sheet||English|
|TEDS - Datenblatt||German|
|TEDS - Prospetto Dati||Italian|
|TEDS - Mounting instructions||English, German|
|TEDS - Bedienungsanleitung||German|
|TEDS - Operating Manual||English|
IEEE Code for TEDS
Conversion template containing the IEEE codes for TEDS of all HBM transducers.
Date: 02/2016 | Size: 2 KB
|Declaration of Conformity|
|1-KAB, K-KAB (cable)- 適合宣言||多言語|
|TEDS - Information||English, German|
|TEDS - Informazione||Italian|
TEDS - Editor
Software to read and write TEDS-data (IEEE 1451.4) into and from 1-wire and 0-wire TEDS-modules and backup it on PC.
Date: 09/2019 | Version: 3.5R43 | Size: 26 MB